Payment & Shipping Terms:
|Model Number:||Gold-Accu Glucose Meter||Test Method:||FAD-GDH System|
|Blood Sample:||Fresh Capillary Whole Blood,venous Whole Blood, Arterial Whole Blood||Calibration Sample:||Venous Plasma|
|Blood Volume:||Less Than 0.8ul||HCT:||20-70%|
|Fast Test Time:||5s Fast Test Time||Memory:||500 Sets Blood Glucose Test Values,100 Sets Control Solution Test Values|
|Expiry Date:||24 Months||Battery Lifetime:||Up To 1000 Times|
Wild hematocrit range Blood glucose machine
system accurency result No coding USB Optional
|Blood sample:||fresh capillary whole blood,venous whole blood, arterial whole blood|
|Calibration sample:||venous plasma|
|Blood volume:||less than 0.8ul|
|Expiry date:||24 months|
|Memory:||500 sets blood glucose test values,100 sets control solution test values|
|Battery lifetime:||1000 tests|
*High quality Blood Glucose Meter
*5 Seconds testing time
*180° Applying Blood
*Strong Anti-interference performance
*More Reliable More professional
*High Accuracy ,long validity
*No coding , easy to use
*Wide HTC range
*Suitable for hospital
Remove a new test strip within the valid expiration from vial. After taking the strip, immediately cover the vial. To avoid losing efficiency. Insert test strip into the strip slot, the test strip is suggested to use up within 3 minutes.
Write the discard date (6 months after first opening the vial) on the vial label when you first open it. Discard the remaining test strips after the discard date.
Test Strips are for single use only. Never reuse a test strip that has had blood applied to it.
In the global market aspect, explore the international market through cooperation of independent brands, develop OEM products, by maintaining good operation ability, and continuously reduce costs in developed countries;, to deliver success in developing countries by product quality and cost performance.
Contact Person: Gao lin